Once these are installed, open your vi. So try the following: Let’s see if I can get you on your way with the following: Go to the block diagram. Start with HPA Init.
|Date Added:||3 October 2017|
|File Size:||59.10 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Hello Joe and Glib, Thank for your help. Of course you can also communicate with your power supply like this. Message 10 of How can i do with these?
Unfortunately, I did not see any examples or “getting started” vi’s for this one. Remote operation of the instrument usually: Message 8 of Right after that you read the answer from your instrument with the GPIB Message 5 of Message 2 of You’ll get there quickly. With the above named VI you first send kind of an initialization to the instrument: Message 3 of Thank you very much for your attention.
Of course you can add some kind of control of the communication if you want. Most Active Software Boards: You can now ask the instrument for the currently gpob data i.
Message 9 of Here is a link for the driver: Look for the appropriate SCPI-commands in your manual. Hope that is of any help for you.
How can I do this? Go to the block diagram.
Let’s see if I can get you on your way with the following: See attached picture for additional functions: You’ve got some code there only for the 3401a the visa referecne is correct and you’re asking for the voltage not current. Now I can do this. You surely know the GPIB-adress of your instrument and the number of your bus while it can sometimes be difficult to get a correct reference.
A GPIB VISA TIME OUT ERRORS | Keysight Community
Before you finish your program you should return your instrument from the remote state in the local state. You have to put together the building blocks to get your measurement. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
Install the driver and follow the steps in my original answer. Did you gpi the driver and install it?
In this way you can communicate with every GPIB-instrument easily without the absolut need of an instrument driver. I don’t understand the attachment.